Do more of the things you love, without worries, thanks to the
Approximately 82.5 x 56 x 21 mm (3.2 x 2.2 x 0.8 inches) without adapter
During operation: 5°C–40°C (41°F–104°F)
During operation: 20%–90%
During operation: 70–106 kPA (700–1060 mbar) (do not exceed 3000 meters [10,000 feet] above sea level)
Minimum = 0.05 U/h, maximum. = 50 U/h.
There are 24 hourly basal rates, adjustable in unit increments of 0.01 (up to 1.00 U/h), 0.05 (up to 10.0 U/h) and 0.1 (up to 50.0 U/h).
Adjustable in 10% increments, 0%–90% (decreases), 110%–500% (increases).
The duration is adjustable in 15-minute intervals, up to a maximum of 12 hours.
The last programmed duration is given by default for the next temporary basal rate change programmed.
400 kPa (4.0 bar), for plastic cartridges and infusion sets
During infusion-set priming and bolus: 0.2 U/sec.
IPX 8—protected against the effects of temporary immersion in water under standardized conditions (60 minutes and 2.5 meters [8 feet])
1 1.5-volt AA alkaline (LR6) or lithium (FR6) battery or 1 1.2-volt rechargeable NiMH AA (HR6) battery.
Batteries should have a minimum capacity of 2500 mAh.
Use only a battery charger officially recommended by the battery manufacturer.
The time and date are safely stored in the memory for about 1 hour after the battery has been removed. Other insulin pump settings (the hourly basal rates, remaining cartridge content, bolus increments and active user menu) and the event memory (bolus history, history of daily insulin totals, temporary basal rate history, and alarm history) are saved, regardless of battery voltage and the time your insulin pump has been without a battery.
Accu-Chek Spirit 3.15 ml cartridge system and Accu-Chek 3.15 ml plastic cartridge with a lure-lock connection. Insuman cartridges*** have been evaluated and approved for use with the Accu-Chek Spirit Combo insulin pump.
The time and date are safely stored in the memory for about 1 hour after the battery has been removed. Other insulin pump settings (the hourly basal rates, remaining cartridge content, bolus increments and active user menu) and the event memory (bolus history, history of daily insulin totals, Temporary Basal Rate history, and alarm history) are saved, regardless of battery voltage and the time your insulin pump has been without a battery.
Accu-Chek infusion sets with a luer-lock connector. (Accu-Chek infusion sets are PVC and latex free.)
Alert system, beeps, information on the display, vibrations, and dual microprocessors.
The insulin pump is controlled by two microprocessors.
The safety concept is based on one processor (supervisor processor) supervising the other (main processor).
Whenever a defect or fault occurs in the main processor, it is identified by the supervisor processor.
The motor is immediately switched off and an “Error E7: ELECTRONIC ERROR” occurs.
Conversely, the main processor can also determine at any time whether the supervisor is working correctly.
The motor also constitutes an important safety component, as the combination of the main and supervisor processor and brushless motor provides the best possible reliability and accuracy in insulin delivery.
*** INSUMAN INFUSAT is a registered trademark of Sanofi S.A.s
94 x 55 x 25 mm
Approximately 103 grams with batteriescell
Colour dot matrix
3 AAA alkaline batteries
See Accu-Chek Aviva test strip package insert
10°C to 50°C with batteries inserted
1000 diary records
Infrared (IR) window to download to information management solutions
7, 14, 30, 60, 90 days